\documentclass[12pt]{article}
\title{BGM3 zero-damp test report}
\author{HERR, vessel: SALLY RIDE}
\date{Release Date: 2017/01/25 18:24:52 UTC}
\begin{document}
\maketitle
Sensor: S224

Software version: 1.2

Port/Pier/Berth: MARFAC

\begin{table}[h]
\centering
\begin{tabular}{ | c | c | }
\hline
Axis & pitch\\ \hline 
Periods & 4\\ \hline 
Test start & 2017/01/25 18:15:39.298\\ \hline 
T1 & 2017/01/25 18:17:52.949\\ \hline 
T2 & 2017/01/25 18:20:07.309\\ \hline 
T3 & 2017/01/25 18:22:21.065\\ \hline 
Test end & 2017/01/25 18:24:35.364\\ \hline 
Tolerance & 5 s\\ \hline 
4T Error & 3.066 s\\ \hline \hline 
\end{tabular}
\caption{Timings}
\end{table}

Action: Performance within specification and tolerance. No action required.

\end{document}

