\documentclass[12pt]{article}
\title{BGM3 zero-damp test report}
\author{Maxwell Hughes, vessel: R/V Sally Ride}
\date{Release Date: 2022/02/25 18:38:09 UTC}
\begin{document}
\maketitle
Sensor: S224

Software version: 1.2

Port/Pier/Berth: MarFac

\begin{table}[h]
\centering
\begin{tabular}{ | c | c | }
\hline
Axis & pitch\\ \hline 
Periods & 4\\ \hline 
Test start & 2022/02/25 18:29:15.873\\ \hline 
T1 & 2022/02/25 18:31:28.420\\ \hline 
T2 & 2022/02/25 18:33:39.663\\ \hline 
T3 & 2022/02/25 18:35:51.533\\ \hline 
Test end & 2022/02/25 18:38:03.440\\ \hline 
Tolerance & 5 s\\ \hline 
4T Error & -5.433 s\\ \hline \hline 
\end{tabular}
\caption{Timings}
\end{table}

Action: Turn potentiometer R65 on Gyro Electronics CPS board 4.1 turns in CCW direction, then retest.

\end{document}

