\documentclass[12pt]{article}
\title{BGM3 zero-damp test report}
\author{Maxwell Hughes, vessel: R/V Sally Ride}
\date{Release Date: 2022/02/25 22:43:27 UTC}
\begin{document}
\maketitle
Sensor: S224

Software version: 1.2

Port/Pier/Berth: MarFac

\begin{table}[h]
\centering
\begin{tabular}{ | c | c | }
\hline
Axis & roll\\ \hline 
Periods & 4\\ \hline 
Test start & 2022/02/25 22:34:29.523\\ \hline 
T1 & 2022/02/25 22:36:43.407\\ \hline 
T2 & 2022/02/25 22:38:57.202\\ \hline 
T3 & 2022/02/25 22:41:10.946\\ \hline 
Test end & 2022/02/25 22:43:24.736\\ \hline 
Tolerance & 5 s\\ \hline 
4T Error & 2.213 s\\ \hline \hline 
\end{tabular}
\caption{Timings}
\end{table}

Action: Performance within specification and tolerance. No action required.

\end{document}

