\documentclass[12pt]{article}
\title{BGM3 tilt test report}
\author{Annie Rosen, vessel: R/V Sally Ride}
\date{Release Date: 2018/07/13 17:21:59 UTC}
\begin{document}
\maketitle
Sensor: S224

Software version: 1.2

Port/Pier/Berth: OSU pier, Newport, OR

\begin{table}[h]
\centering
\begin{tabular}{ | c | c | }
\hline
Axis & roll\\ \hline 
Filtered gravity 1 & 980562.39\\ \hline 
Filtered gravity 2 & 980562.37\\ \hline 
Misalignment & 0.0091478265 mrad \\ \hline 
Tolerance & 0.7mrad \\ \hline \hline
\end{tabular}
\caption{Gravity readings and misalignment}
\end{table}

Action: Performance within specification and tolerance. No action required.

\end{document}

