\documentclass[12pt]{article}
\title{BGM3 tilt test report}
\author{Maxwell Hughes, vessel: R/V Sally Ride}
\date{Release Date: 2022/02/25 17:25:28 UTC}
\begin{document}
\maketitle
Sensor: S224

Software version: 1.2

Port/Pier/Berth: MarFac

\begin{table}[h]
\centering
\begin{tabular}{ | c | c | }
\hline
Axis & roll\\ \hline 
Filtered gravity 1 & 979269.2\\ \hline 
Filtered gravity 2 & 979269.95\\ \hline 
Misalignment & -0.27156492 mrad \\ \hline 
Tolerance & 0.7mrad \\ \hline \hline
\end{tabular}
\caption{Gravity readings and misalignment}
\end{table}

Action: Performance within specification and tolerance. No action required.

\end{document}

