\documentclass[12pt]{article}
\title{BGM3 ship-to-shore gravity tie report}
\author{Maxwell Hughes, vessel: R/V Sally Ride}
\date{Release Date: 2022/02/28 18:59:12 UTC}
\begin{document}
\maketitle
Sensor: S224

Software version: 1.2

Port/Pier/Berth: MarFac San Diego

\begin{table}[h]
\centering
\begin{tabular}{ | c | c | }
\hline
Gravity station number & 0032.84\\ \hline 
Station name & Nimitz Marine Facility\\ \hline 
mGal at pier & 979535.93\\ \hline 
Tie start time UTC & 2022/02/28 17:53:57.592\\ \hline 
Samples used & 3600\\ \hline 
Land tie used & No\\ \hline 
Water height to pier 1 & 7 ft 11 in \\ \hline 
Water height to pier 2 & 8 ft 8 in \\ \hline 
Water height to pier 3 & 9 ft 8 in \\ \hline 
Average of filtered counts & 24927.189354056\\ \hline 
Filter length & 361\\ \hline 
Scale factor & 4.98247686\\ \hline \hline 
\textbf{NEW BIAS} & \textbf{855337.61}\\ \hline 
\end{tabular}
\caption{Gravity tie information}
\end{table}

\end{document}

