\documentclass[12pt]{article}
\title{BGM3 ship-to-shore gravity tie report}
\author{H. Johnson, M. Thompson, vessel: R/V Sally Ride}
\date{Release Date: 2022/06/01 18:53:18 UTC}
\begin{document}
\maketitle
Sensor: S224

Software version: 1.2

Port/Pier/Berth: MarFac 

\begin{table}[h]
\centering
\begin{tabular}{ | c | c | }
\hline
Gravity station number & NMF-AG 2017\\ \hline 
Station name & Nimitz Marine Facility - AG\\ \hline 
mGal at pier & 979535.93\\ \hline 
Tie start time UTC & 2022/05/31 22:35:54.589\\ \hline 
Samples used & 3600\\ \hline 
Land tie used & No\\ \hline 
Water height to pier 1 & 9 ft 11 in \\ \hline 
Water height to pier 2 & 9 ft 9 in \\ \hline 
Water height to pier 3 & 9 ft 8 in \\ \hline 
Average of filtered counts & 24927.207635444\\ \hline 
Filter length & 361\\ \hline 
Scale factor & 4.98247686\\ \hline \hline 
\textbf{NEW BIAS} & \textbf{855337.61}\\ \hline 
\end{tabular}
\caption{Gravity tie information}
\end{table}

\end{document}

