\documentclass[12pt]{article}
\title{BGM3 periodic checks report}
\author{herr, vessel: sally ride}
\date{Release Date: 2017/01/25 17:34:11 UTC}
\begin{document}
\maketitle
Sensor: S224

Software version: 1.2

Port/Pier/Berth: marfac san diego

Wind sea: 0 degree: no wave, calm(glassy)

Swell: 0 degree: no swell

\begin{table}[h]
\centering
\begin{tabular}{ | c | c || c | c || c | }
\hline
Test Point & Value & Min Spec & Max Spec & In Spec\\ \hline \hline
Sensor TP 1 & 30.01 & 29 & 31 & OK\\ \hline 
Sensor TP 2 & 27.66 & 26 & 28 & OK\\ \hline 
Sensor TP 3 & 18.12 & 18.5 & 19.5 & OUT\\ \hline 
Sensor TP 4 & 17.98 & 17.1 & 18.9 & OK\\ \hline 
Sensor TP 5 & 14.99 & 14.2 & 15.8 & OK\\ \hline 
Sensor TP 6 & -15 & -15.8 & -14.2 & OK\\ \hline 
Sensor TP 7 & 0 & -100 & 100 & OK\\ \hline 
Sensor TP 8 & 4.99 & 4.75 & 5.25 & OK\\ \hline 
Sensor TP 9 & 14.23 & 0 & 15 & OK\\ \hline 
Sensor TP 10 & 18.4 & 17.1 & 18.9 & OK\\ \hline 
Sensor TP 11 & 6.43 & 6 & 6.8 & OK\\ \hline 
Sensor TP 12 & 0 & -2 & 2 & OK\\ \hline 
Sensor TP 13 & 0 & 0 & 15 & OK\\ \hline 
Sensor TP 14 & -18.68 & -18.9 & -17.1 & OK\\ \hline 
Sensor TP 15 & -6.9 & -7.4 & -6.6 & OK\\ \hline 
Sensor TP 16 & 0 & -0.001 & 0.001 & OK\\ \hline 
\end{tabular}
\caption{Sensor test points}
\end{table}

\begin{table}[h]
\centering
\begin{tabular}{ | c | c || c | c || c | }
\hline
Test Point & Value & Min Spec & Max Spec & In Spec \\ \hline \hline
Control Power Supply TP 1 & 14.74 & 14 & 16 & OK\\ \hline 
Control Power Supply TP 2 & 10.16 & 9 & 11 & OK\\ \hline 
Control Power Supply TP 3 & 1.02 & 0.9 & 1.1 & OK\\ \hline 
Control Power Supply TP 4 & 0.01 & -1 & 1 & OK\\ \hline 
Control Power Supply TP 5 & 0.02 & -1 & 1 & OK\\ \hline 
Control Power Supply TP 6 & 1.93 & -3 & 3 & OK\\ \hline 
Control Power Supply TP 7 & -2.57 & -3 & 3 & OK\\ \hline 
Control Power Supply TP 8 & 0.03 & -0.2 & 0.2 & OK\\ \hline 
Control Power Supply TP 9 & 0.03 & -0.2 & 0.2 & OK\\ \hline 
\end{tabular}
\caption{CPS test points}
\end{table}

\begin{table}[h]
\centering
\begin{tabular}{ | c | c | }
\hline
Observation & Value \\ \hline \hline
Max accel oven & 99.99\\ \hline 
Accel oven cycling & Yes\\ \hline 
Max prc oven & 40\\ \hline 
Prc oven cycling & Yes\\ \hline 
Max bptc oven & 0.7\\ \hline 
Bptc oven cycling & Yes\\ \hline 
Battery volt & 27\\ \hline 
Elex volt & 28.5\\ \hline 
Elex current & 0.18\\ \hline 
Charge current & 0.1\\ \hline 
\end{tabular}
\caption{Observations}
\end{table}

\end{document}

